Technical video for the Rudolph DDM 2911 Density Meter. This video details the procedure required to clean the Density Meter prior to or after making a liquid density measurement.
The video covers using a syringe to inject cleaning solvent, drying agent, capturing waste, and drying the instrument's U-Tube with the on-board air pump. Using VideoView TM and simple test measurement the user can verify that the instrument is clean, dry, and ready for a new liquid density measurement.
For additional information visit our website at http://www.rudolphresearch.com